Gandus Saldatrici Indicator BlueTest.it is a new generation Indicator test for routing testing of your sealing devices. Specially developed to check the integrity of seals produced with your sealers. Suitable for Paper and Tyvek®backed laminated pouches.
It allows a safe, easy and quick ink test according to ISO 11607-1-2 and ASTM1929-12 .
Eliminates any risk of leakage or migration of the methylene blue dye
No damage to seal. Easy to prepare and easy to check the result
Check after 5 seconds if the seal made with your device is perfect, secure and safe
How to use BlueTest.it
Put the BlueTest.it into the pouch or sleeve
Seal the pouch or sleeve
Press by hand in the direction of the arrow
Wait for 20 second – no longer – (if testing all 4 seals, allow 5 seconds link contact on each)
After 20 seconds (no longer) check the integrity of the seal area and that the blue ink does not enter any part of the seal (eg channels)
Quantity / box
BlueTest.it indicator Test Paper Packing
BlueTest.it indicator Test Tyvek® Packing
Test system for daily testing of sealing device and for Operational Qualification according to EN ISO 11607-2:2006 process validation.
Guidelines of the WFHSS advise to test each medical heat sealer at the beginning of the day after the heating up.
Based upon the test the heat sealer can be released for the day.
Gandus Saldatrici Seal CheckIT is the perfect way to comply to the ISO 11607 – 2 requirements for Operational Qualification (OQ) and daily testing of your Medical Heat sealers
Is the sealing temperature defined and correct?
Are speed and force the most suitable for that kind of product?
Did the manufacturer indicate these parameters inside the specs?
How does it works?
Place in between the layers of medical pouches with face up to the laminate side of the pouch. The combination is then heat-sealed. By its unique printed pattern of the black area the inspection of the seal and the detection of minor or bigger problems are easily detected.
Sealing temperature too low
Sealing pressure too low
Interruptions in the seal
Compliance to standards
Procedure: ISO 11607 – 2
Materials: ISO 868 – 3
Technical details each test
Dimensions: 175 x 75 mm. Materials:
Paper: 60 gr/m², Medical Grade
Inks: waterbsed, non solvent, non-toxic, non-heavy metals Box: each box is made of carton (180 x 80 x 25 mm.) and contains 250 pcs.